If you like the ChemWiki, please "like" us on facebook, "share" us on Google+, or "tweet" about the project.
Vacuum equipment is used to generate, maintain, and manipulate pressures below that of the ambient atmosphere. Many common lab procedures require vacuum conditions, such as inert gas purging, cannulation, and solvent evaporation. Vacuum equipment often requires special care to maintain.
Higher quality vacuum contains less matter, and therefore a lower pressure. A commonly used convention is:
A Schlenk line, often referred to as a vacuum manifold, is composed of a vacuum line connected by several valves to an inert gas line and a port. The valve position connects the port to either the vacuum line or the inert gas line. The inert gas line often has an oil-filled bubbler at its outlet to prevent atmospheric contamination.
A McLeod Gauge is an instrument for measuring pressure in high-vacuum systems. It is filled (typically) with mercury. When pressure is not being measured, the gauge should be held in the "evacuating" position; the mercury rests in a chamber connected to two capillary tubes, which should be (just past) horizontal. To make a measurement, the gauge should be connected the line and opened to vacuum. Pressure should be allowed to equilibrate for a couple of minutes. Then the gauge should be rotated so that the capillaries are vertical (the "measuring" position); the mercury should flow from the chamber into the capillaries. One of these capillaries is open to the rest of the system, while the other is closed. The difference between the levels of the mercury when these are vertical gives the pressure of the system (the instrument should have its own scale, which should be zeroed to the level of mercury in the open capillary). For accurate measurements, the rotation from evacuating to measuring positions should be slow, to prevent discontinuities in the mercury.
Before disconnecting the gauge, it should be returned to the evacuating position and returned, slowly, to atmospheric pressure. Disconnecting the system while it is still under vacuum can cause splattering of mercury, potentially out of the gauge. Returning it to atmospheric pressure while it is in the measuring position can cause mercury to get stuck in the closed capillary.
Care should be used to avoid venting harmful vapors into the lab atmosphere. Pump exhaust should be either vented into a fume hood, or fitted with an appropriate scrubber or filter. When broken, evacuated glassware shatters and violently implodes, sending fragments flying at high velocity. Inspect your glassware for cracks and flaws before applying vacuum.
An NSF funded Project
By STEMWiki Hyperlibrary